This project is about hacking the Voltcraft CO-100 CO2-sensor, such that we can read the exact ppm value as displayed on the LCD.
This particular CO2-sensor is present in the klusbunker at RevSpace and is currently used to control the ventilation in a crude manner (by monitoring the warning LEDs on the display module). Having the CO2 ppm value available as a number allows for nice things such as logging the levels over time, announce them on IRC, show them on the LedBanner , etc.
The user manual of the Voltcraft CO-100 says "Attention! The RJ45 connection (see chapter 7, item „K“) must not be used. The connection is only intended for the manufacturer". Of course, a claim like that can only be interpreted as a challenge! :)
The CO2 sensor inside the CO-100 (in the left of the picture) has a sticker saying ZGw063RY. Googling for this number reveals a CO2 module that looks just like the Voltcraft CO-100, so it appears that the CO-100 is basically just a rebranded ZyAura ZGw063RY module.
The CO-100 seems to miss a bunch of components that can be mounted on the PCB, close to the RJ45 connection (possibly an RS232 chip with step-up capacitors).
The CO2 sensor in the CO-100 looks very much like the ZyAura ZG-01 module. According to the description, this uses a proprietary ZyAura digital protocol.
The ZG-01 sensor apparently uses a SPI protocol.
Other sensors on the market:
Possible further investigation:
- Sniff the ZG01 SPI connection, e.g. using a bus pirate or an Arduino
- Check if the empty footprint on the PCB matches a max232